International
Tables for Crystallography Volume C Mathematical, physical and chemical tables Edited by E. Prince © International Union of Crystallography 2006 
International Tables for Crystallography (2006). Vol. C, ch. 5.2, p. 498

Since about 1930, it has been claimed that the lattice parameters of cubic substances could be measured within one part in 50 000. Precision (that is, reproducibility of measurements by one technique within one laboratory) of this order is achieved, but accuracy (agreement between determinations by different techniques or by the same technique in different laboratories) is lower. The IUCr latticeparameter project (Parrish, 1960) showed a standard deviation of 1 in 30 000 in interlaboratory comparisons, with some outlying values differing from the mean by one or two parts in 10 000. At that time, therefore, precision was considerably better than accuracy (absence of significant remanent systematic error). Testing for remanent systematic error is thus valuable as an occasional test of methodology, though not undertaken as routine. The principle is outlined here, and more details are given in Chapters 8.4 and 8.5 .
When refinement of parameters is performed by least squares, weighted in accordance with the reciprocal of the estimated variance, the expected value of the weighted sum of squares is where n is the number of terms summed and p is the number of parameters determined. The standard deviation of the sum S is expected to be approximately (Wilson, 1980), so that if the actual value of S exceeds (where k = 2 or 3), one can reasonably conclude that there are defects in the model (remanent systematic errors). If S is less than this value, one can reasonably conclude that any defects in the model (systematic errors) are at worst of the same order of magnitude as the statistical fluctuations; the sensitivity of the test increases rather slowly with n − p. The method was advocated by Beu and his collaborators (Beu, Musil & Whitney, 1962, 1963; Beu, 1964; Beu & Whitney, 1967; Langford, Pike & Beu, 1964; see also Mitra, Ahmed & Das Gupta, 1985) because tests of the hypothesis `no remaining systematic error' based on likelihood were available; they assumed a normal distribution of errors, possibly without realizing, and certainly without emphasizing, that the method was then equivalent to least squares. Their application of the method to testing for remanent systematic error in latticeparameter determination was successful: the aberrations of the counter diffractometer were found to be adequately accounted for: additional aberrations were found for the Bond method (see Chapter 5.3 ); Boom (1966) used it in testing the accuracy of the Debye–Scherrer method.
In statistical literature, the weighted sum of squares S is often called the scaled deviance, and is called the excess. The test for the absence of significant systematic error is then that the excess should be less than three.
References
Beu, K. E. (1964). The evaluation of centroid lattice parameter data for tungsten by the likelihood ratio method. Acta Cryst. 17, 1149–1164.Beu, K. E., Musil, F. J. & Whitney, D. R. (1962). Precise and accurate lattice parameters by film powder methods. I. The likelihood ratio method. Acta Cryst. 15, 1292–1301.
Beu, K. E., Musil, F. J. & Whitney, D. R. (1963). The likelihood ratio method for the precise and accurate determination of lattice parameters for tetragonal and hexagonal crystals. Acta Cryst. 16, 1241–1242.
Beu, K. E. & Whitney, D. R. (1967). Further developments in a likelihood ratio method for the precise and accurate determination of lattice parameters. Acta Cryst. 22, 932–933.
Boom, G. (1966). Accurate lattice parameters and the LPC method. Groningen: van Denderen.
Langford, J. I., Pike, E. R. & Beu, K. E. (1964). Precise and accurate lattice parameters by film powder methods. IV. Theoretical calculation of axial (vertical) divergence profiles, centroid shifts, and variances for cylindrical powder diffraction cameras. Acta Cryst. 17, 645–651.
Mitra, G. B., Ahmed, A. & Das Gupta, P. (1985). Precise and accurate estimation of crystallographic parameters by maximumlikelihood and min–max methods. Structure and statistics in crystallography, edited by A. J. C. Wilson, pp. 151–181. Guilderland, NY: Adenine Press.
Parrish, W. (1960). Results of the IUCr precision latticeparameter project. Acta Cryst. 13, 838–850.
Wilson, A. J. C. (1980). Accuracy in methods of latticeparameter measurement. Natl Bur. Stand. (US) Spec. Publ. No. 567, pp. 325–351.